Microelectronic stimulator array for stimulating nerve tissue

ABSTRACT

The retinal prosthesis test device is comprised of a thin wafer of glass made from nanochannel glass (NGC) with very small channels perpendicular to the plane of the wafer filled with an electrical conductor forming microwires. One surface of the glass is ground to a spherical shape consistent with the radius of curvature of the inside of the retina. The NGC is hybridized to a silicon de-multiplexer and a video image is serially input to a narrow , flexible micro-cable and read into a 2-D array of unit cells in a pixel-by-pixel manner which samples the analog video input and stores the value as a charge on a MOS capacitor. After all unit cells have been loaded with the pixel values for the current frame, a biphasic pulse is sent to each unit cell which modulates the pulse in proportion to the pixel value stored therein. Because the biphasic pulses flow in parallel to each unit cell from a global external connection, the adjacent retinal neurons are all stimulated simultaneously, analogous to image photons stimulating photoreceptors in a normal retina. A permanent retinal implant device uses a NGC array hybridized to a silicon chip, the image is simultaneously generated within each cell through a photon-to-electron conversion using a silicon photodiode. The photons propagate directly through into the backside of the device. Electrical power and any control signals are transmitted through an inductively driven coil or antenna on the chip. The device collects the charge in storage capacitors via the photon-to-electron conversion process, stimulates the neural tissue with biphasic pulses in proportion to the stored charges, and resets the storage capacitors to repeat the process.

BACKGROUND OF THE INVENTION

[0001] 1. Field of the Invention

[0002] This invention deals generally with neural prosthesis,specifically the concept of achieving a retinal prosthesis for blindpatients through the creation of an electrical interface between ahigh-density electrode array and the curved surface of the retina.

[0003] 2. Description of the Related Prior Art

[0004] There is a great deal of recent interest in the area of neuralprosthesis, specifically the concept of achieving a retinal prosthesisfor blind patients has been hypothesized by a number of researchers andis an active area of medical research. In a normal eye, in a basicconcept 10, FIGS. 1a and 1 b shows a ray trace of two photons 12 focusedon a retina 21. Note that the incoming photons 12 pass through severallayers of transparent retinal cells 16 and 18 before being absorbed bythe photoreceptors 22. In a damaged eye, a retinal prosthesis device 24,as shown in FIGS. 1c and 1 d, is positioned against the retina 21. Inthis case, the photons 12 are absorbed by a microelectronic stimulatingarray or device 26 that is hybridized to a glass piece 28 containing anembedded array of microwires. The glass has a curved surface thatconforms to the inner radius of the retina 21. The microelectronicimaging device 26 is made of thin silicon containing very large scaleintegrated (VSLI) circuitry and photon detectors that convert theincident photons 12 to an electronic charge. The charge is thenconverted to a proportional amount of electronic current which is inputto the nearby retinal cell layer 18. The cells fire and a signal is sentto the optic nerve 28.

[0005] A typical retinal prosthesis device combines two technologies:first, nanochannel glass (NGC) electrode arrays and secondly atwo-dimensional (2-D) multiplexer array. NGC technology employs fiberoptic fabrication techniques to produce thin wafers of glass with verysmall channels perpendicular to the plane of the wafer. Typical NGCwafers that will be required for retinal prosthesis devices are severalmillimeters in diameter and can contain millions of channels withchannel diameters on the order of one micron. The channels are filledwith a good electrical conductor and one surface of the glass is groundto a spherical shape consistent with the radius of curvature of theinside of the retina. The electrical conductors on the curved surfaceshould protrude slightly to form efficient electrodes.

[0006] The 2-D multiplexer array is similar to infrared focal planearray (IRFPA) multiplexers that are microelectronic devices fabricatedat silicon foundries. An IRFPA multiplexer is a 2-D array that reads outthe infrared (IR) image captured by a complimentary detector array thatconverts photons into electrical charge. The charge is integrated andstored in each unit cell for a few milliseconds. The full image is thenmultiplexed off the array at frame rates compatible with commercialvideo. For a retinal prosthesis test device that obtains its input imagefrom an external camera, the process is essentially reversed and thedevice acts as a de-multiplexer. That is, the prosthesis devices willperform de-multiplexing operations, but will be referred to here simplyas a multiplexer.

[0007] The basic concept is straightforward: visual images can beproduced in the brain by electrical stimulation of retinal cells.Two-dimensional arrays of retinal cells, such as ganglion or bipolarcells, can be stimulated using two-dimensional arrays of electricalimpulses with the spatial form of an image. The axons of the ganglioncells then transmit the image through the optic nerve and on to thevisual cortex. This is in lieu of the normal photo-transduction processthat occurs in a healthy retina. In approximately 90 percent of blindpatients, the photoreceptors are diseased, but the other retinal layersare still responsive to electrical stimulation.

[0008] Experimental test procedures, such as shown in FIG. 2, usestandard retinal surgical techniques performed in an operating roomenvironment by an ophtalmologist. It is necessary that the patient beadministered local anesthesia rather than general anesthesia so thatvisual perceptions can be orally recorded during the procedure.

[0009] There are a number of technical issues to be addressed indesigning and fabricating a retinal prosthesis device, particularly ifthe device is to generate a high resolution image. First, there is theissue of creating an electrical interface between the high-densityelectrode array and the curved surface of the retina. The electrodearray must have a spherical, convexed shape in order to conform to thespherical concaved surface of the retina. The electrode array must bebio-compatible and safe for permanent implantation. Second, theelectrical stimulation pulse shapes and repetition rates, whilegenerally well known, may need to be optimized for each individualrecipient of a prosthesis device. The pulse amplitude is of coursemodulated within the retina to be proportional to the pixel value.Third, direct electrical stimulation of the ganglion cells precludescertain image processing functions that normally would have occurred inearlier layers of the retina. Therefore, computationally based imagepreprocessing operations may need to be performed on the image beforestimulation of the retina. Fourth, supplying power to a permanentimplant will need to be engineered in a manner such that there are nowires or cables through the eye wall. Fifth, because a normal retinaprocesses image information created by the photoreceptors in asimultaneous manner, it is assumed that a prosthesis device shouldsimilarly excite retinal cells in a simultaneous manner, as opposed tosequential raster scan that might cause synchronicity problemsdownstream in the lateral geniculate nucleus (LGN) or visual cortex.

SUMMARY OF THE INVENTION

[0010] An object of this invention is to provide a device for achievinga retinal prosthesis for blind patients.

[0011] Another object of this invention is to provide a retinalprosthesis test device for providing visual images to the brain duringacute human experiments to achieve electrical stimulation of the retinatissue.

[0012] Another object of this invention is to provide a device forimplant into the human eye that will allow electrical stimulation of theretinal or any neural tissue so as to provide visual images to thebrain.

[0013] These and other objects are accomplished by the retinalprosthesis test device and retinal implant device comprising two basictechnologies—nanochannel glass (NGC) electrode arrays and infrared focalplane array (IRFPA) multiplexers. In the retinal prosthesis test device,the device is positioned against the retina using standard retinalsurgical techniques in an operating room environment. The device iscomprised of a thin wafer of glass (NGC) with very small channelsperpendicular to the plane of the wafer. The channels are filled with agood electrical conductor forming microwires with one surface of theglass being ground to a spherical shape consistent with the radius ofcurvature of the inside of the retina. Electrical conductors protrudeslightly from the NGC on the curved surface to form electrodes. The NGCis hybridized to a silicon IRFPA multiplexer using indium bump bonds. Animage is serially input into the multiplexer via a very narrow, flexiblemicro-cable. The multiplexer is mounted on a ceramic carrier such thatinterconnecting bond pads on each are in close proximity to one another.A video image is read into each of the unit cells on the multiplexer inpixel-by-pixel manner. Discrete samples of the analog video are inputand stored as electrical charge on a MOS capacitor. After all unit cellshave been loaded with the pixel values for the current frame, a biphasicpulse is sent through each unit cell and into the corresponding area ofthe retina. The biphase pulse is modulated in proportion to the pixelvalue stored therein. Because the biphasic pulse flows in parallel froma global external connection, the adjacent retinal neurons are allstimulated simultaneously, analogous to image photons stimulatingphotoreceptors in a normal retina.

[0014] A permanent retinal implant device uses an NGC array hybridizedto a silicon chip in an identical manner to the retinal prosthesis testdevice, however, the image is no longer multiplexed onto the chipthrough a wire from an external camera, but instead, the image issimultaneously generated within each cell through a photon-to-electronconversion using a silicon photodiode. The photons propagate directlyinto the backside of the device. Electrical power and any controlsignals are transmitted through an inductively driven coil or antenna onthe chip. The device collects the charge in storage capacitors via thephoton-to-electron conversion process, stimulates the neural tissue withbiphasic pulses in proportion to the stored charges, and resets thestorage capacitors to repeat the process.

BRIEF DESCRIPTION OF THE DRAWINGS

[0015]FIG. 1a shows a general diagram of a ray trace of photons incidenton a retina without a prosthesis device (a normal eye).

[0016]FIG. 1b shows an interior view of a ray trace of photons incidenton a retina without a prosthesis device (a normal eye).

[0017]FIG. 1c shows a general diagram of a ray trace of photons incidenton a retina with a prosthesis device (prosthesis device positionedagainst the retina).

[0018]FIG. 1d shows an interior view of a ray trace of photons incidenton a retina with a prosthesis device (prosthesis device positionedagainst the retina).

[0019]FIG. 2 shows a retinal prosthesis test device positioned against aretina.

[0020]FIG. 3a shows a side-view of a fully packaged retinal prosthesistest device.

[0021]FIG. 3b shows an enlarged view of a nano-channel glass (NCG)electrode array.

[0022]FIG. 4 shows a conceptual layout (floor plan) of a silicon chipfor the retinal test prosthesis.

[0023]FIG. 5 shows a conceptual design of a unit cell for the retinalprosthesis test device showing the external inputs from off-chip.

[0024]FIG. 6 shows a block diagram of ancillary electronics for theretinal prosthesis test device.

[0025]FIG. 7 shows a side-view of a permanent implant device.

[0026]FIG. 8 shows a conceptual design of a unit cell for a permanentimplant device.

[0027]FIG. 9 shows a conceptual layout (floor plan) of a silicon chipfor a permanent implant device.

DESCRIPTION OF THE PREFERRED EMBODIMENT

[0028] In the preferred embodiment of a retinal prosthesis test deviceutilizing a microelectronic stimulator array 30, as shown in FIG. 3a,the nanochannel glass (NGC) electrode arrays 32 is hybridized to siliconmultiplexer 34 using indium bump bonds, a technique from infrared (IR)focal plane array (IRFPA) multiplexers. An image is serially input ontothe multiplexer 34 via a very narrow, flexible micro-cable 36. Themicro-cable 36 is approximately six inches in length and is custom madeusing gold leads patterned on polyimide strips. A ceramic carrier 38with gold-filled via holes 42 provides a mechanically convenient meansof routing interconnects from the top-side 44 of the ceramic carrier tothe back-side 46. By designing the ceramic carrier such that thetop-side contacts to the via-holes 44 are in close proximity to bondpads 48 on the silicon multiplexer 34, the interconnection may be madewith conventional tab-bonds 52 (thin gold ribbons fused to interconnectswith mechanical pressure as is common practice in the microelectronicsindustry and is well known to those skilled in the art. This keeps allof the interconnects 52 from protruding above the spherical curvedenvelop defined by the polished NCG 32 and therefore prevents damage tothe tab bond interconnects 52 or to a patients retina 54. A criticalissue for any neural prosthesis device is biocompatibility and safety.Because the duration of any tests with the retinal prosthesis testdevice 30 are very short (less than an hour), biocompatibility issuesare primarily reduced to acute effects of the testing and need notaddress the more difficult chronic issues that arise with permanentimplants. Note that the surface of the packaging shown in FIG. 3aconsists only of glass 32, platinum (Pt) electrodes 56, and siliconencapsulation 58. However, as with any medical instrumentation, a majorsafety issue is electrical shock hazard. Note that the purpose of thedevice is to provide minimal electrical stimulation of retinal tissueusing very low voltages and the smallest current possible, i.e.,preferably about one volt and 1 μA per unit cell for about 1 millisecond(every frame at a frame rate of 60 Hz). To protect a patient from anyelectrical shock, the patient is isolate from high voltages usingoptocouplers (not shown) which are powered by low voltage electricalbatteries (not shown), a technique well known to those skilled in theart.

[0029] Specific requirements for the NCG 32 are that the channels 56making up the NCG 32, as shown in FIG. 3b, be small enough so that manymicrowires can be connected to each unit cell of the multiplexer array.This is for redundancy and to help simplify the hybridization alignment.If the NCG wires were to approach the size of the unit cell, then aone-to-one alignment and hybridization would be required. This would bevery problematic, because of irregularities in the NCG periodicity andthe possibility of shorting nearest neighbor cells (not shown). On theother hand, very narrow channels 56 imply very high length-to-widthaspect ratios of the channels 56 in the NCG 32. This makes it difficultto fabricate large area NCG 32 samples with the proper thickness. It isexpected that a reasonable design size for the channels 56 should be onthe order of a micron.

[0030] The NCG channels 56 must be filled with a high conductivitymaterial 62, such as platinum, to create microwires. Fabrication of themicrowires can be performed using electrodeposition or infusion ofmolten metal under pressure, techniques that are well known to thoseskilled in the art. After the channels 56 have been filled withconductive material and the continuity of the microwires has beenconfirmed, one side of the glass 32 must be polished to create aspherical surface 64. This is accomplished by carefully grinding andpolishing of the glass/metal composite. The radius of curvature isnominally half an inch in order to provide a conformal fit against theinside of the retina 54. This is critically important as it allowspositioning of the high-density electrodes in the NCG 32 against theretinal 54 tissue. The polishing process will create slightly recessedmicrowires with respect to the curved NCG 32 surface. This is becausethe metal is softer than the glass. Therefore further processing may benecessary to create electrodes that protrude slightly above the curvedsurface 64 of the NCG 32. In preparation for hybridizing the NCG 32 tothe multiplexer 34, or the microwires may be hybridized directly to theindium bumps 66 on the multiplexer 34 or the glass is etched so themicrowires protrude slightly from the NCG 32. This is similar to themanner used to form the protruding electrodes on the curved side 64 ofthe NCG 32.

[0031] A conceptual layout of the multiplexer 34 is shown in FIG. 4. Thesilicon multiplexer 34 performs several operations in a sequentialorder. During the first step, an image is read onto the multiplexer 34,pixel-by-pixel to each unit cell 72. The row shift register 74 andcolumn shift register 76 control the routing into each unit cell 72. Thediscrete samples of analog video are input and stored as charge on MOScapacitor. This operation occurs every 60^(th) of a second in a mannercompatible with a RS-170 television format allowing the use of the testprosthesis 30 with standard video equipment. A multiplexer 34 that has aread-on and read-off capability has several input signals including apixel clock, start-of-frame clock, bias voltage, ground, and analoginput (RS-170). A digital electronics block 78 is of major importancebecause it generates switching pulses that routes image data into theunit cells 72 by controlling the row shift register 74 and the columnshift register 76. Without the on-chip digital electronics 78, theremight be a dozen or more clocks that would need to be input to thedevice. That would make the cable 27 from external drive electronics 25through the eye wall 29 (as shown in FIG. 2) much larger and morecumbersome. The use of IRFPA multiplexer technology greatly simplifiescable 27 problems through the eye wall 29.

[0032] Referring again to FIG. 4, after all the unit cells 72 have beenloaded with the pixel values for the current frame, the next step is tosend a biphasic pulse to that unit cell 72 which in turn is modulated inproportion to the pixel value stored in each unit cell 72. Because thebiphasic pulse flows in parallel from a global external connection, theadjacent retinal neurons are all stimulated simultaneously. This is animportant feature of the design because it is synchronistic actionanalogous to imaged photons stimulating photoreceptors in a normalretina. Finally, the electrodes are all connected to ground to preventany possible charge build up at the electrode-neuron interface.

[0033] There are several important consideration in designing a devicethat performs all these operations successfully. First the multiplexer32 operation should be designed with many of the requirements that existfor an IRFPA, for example, good uniformity, low noise, and high dynamicrange. Of course, the retinal prosthesis test device 30 moves image datain the opposite direction than an IRFPA multiplexer, that is, image datamoves onto the device rather than off the device, but otherwise thespecifications are analogous. FIG. 5 shows a generic design for a unitcell 72. Note that the unit cell 72 stores the pixel value and then usesit to modulate the biphasic pulse 82 that is input to the retinal tissue54 through the NCG 32. Note that the biphasic pulse 82 and the imagedata 84 are both generated off-chip. This allows for greater flexibilityduring human testing as any image sequence can be input and combinedwith any shape of biphasic pulse 82. The switch 86 at the bottom of FIG.5 provides the capability to connect the retinal tissue 54 to ground 88to avoid any possibility of charge build-up.

[0034] Referring to FIG. 6, the operation of the retinal prosthesis testdevice 30 during acute experiments is controlled and powered by externalancillary electronics 25. The input signal is an image sequence at datarates fast enough to achieve 60 frames per second. The multiplexer 34 isdesigned to sample the multiplexed input signal in a manner compatiblewith the RS-170 format. This allows the retinal prosthesis test device30 to be interfaced directly with any standard video camera. Thisincludes the use of a computer which stores digital imagery and candisplay sequential fields at a 60 Hz rate (RS-170 interlaces two fieldsper frame at a rate of 30 frames per second). The actual control of themicroelectronic multiplexer 34 is done with precisely timed pulsesgenerated by a set of signal clocking boards 96 in a manner similar tothat used in typical IRFPA's. The sync pulse generator 92 is used tosynchronize the RS-170 signal 91 with the clocking pulses 96. Basically,the sync pulse generator 92 detects the beginning of each RS-170 fieldand then sends a corresponding pulse to the drive electronics 94 thattriggers the clocking signals 96 required to control each field of theimage data input to the multiplexer 34, synchronizaton of the pulses canbe monitored with an oscilloscope 93. The isolated breakout box 98electrically isolates the human subject from high voltage powersupplies. The box 98 contains opto-couplers that isolate the clock 96and biphasic pulse signals 82 and low voltage batteries supplying biaspotentials 102.

[0035] The biphasic pulses 82 used to stimulate the retinal tissue 54may be programmable such that any pulse shapes can be tested. This hasseveral important implications for the development process. First,because the input impedance to the retinal tissue 54 has both aresistive and capacitive reactance associated with it, a square wavevoltage pulse will not produce the desired square wave current pulse.Neurobiologists tend to favor square wave current pulses to achieveefficient neural stimulation. With knowledge of the output impedance atthe electrode-retina interface, a voltage shape can be computed thatwill provide a square wave current pulse, thus providing efficientstimulation. Second, there is evidence that various layers of the retina54 can be stimulated with different shaped pulses—probably because oftheir differing frequency responses. Specifically, it is expected thateither the ganglion or bipolar cells can be selectively stimulated.Stimulating the bipolar cells instead of the ganglion cells has theadvantage of reaching more deeply into the retina 54, allowing a morenatural form of stimulation.

[0036] Direct electrical stimulation of the ganglion cells precludescertain processing functions that normally would have occurred in theearlier layers of the retina 54. Therefore, it may be necessary toperform certain functions on the incoming imagery before stimulation tocompensate for the missing processing. Unfortunately, a detailed modelof human retinal functions has never been confirmed. Nevertheless, ithas been shown that many intracellular recordings from the retinas ofrabbits are very similar in mammailian vertebra species in general.

[0037] In another preferred embodiment, a permanent implant device 40,as shown in FIG. 7, that is fully self-contained and responds toincident photons naturally imaged through the lens of the eye, similarto that shown in FIGS. 1c and 1 d, is taught. The device 40 issurgically implanted in a patients eye and has with no externalconnections passing through the eye wall. The basic design of thisdevice 40 is based extensively on the retinal prosthesis test device 30taught above. Specifically, the permanent implant device 40 would use aNCG array 102 hybridized to a silicon chip 104 in an identical manner tothe retinal prosthesis test device 30. However, the unit cell 106circuitry is redesigned because the image is no longer being multiplexedonto the chip through a cable from an external camera, but instead, theimage is simultaneously generated within each unit cell 106 through aphoton-to-electron conversion using a silicon photodiode 108, as shownconceptually in FIG. 8. The photons 112 propagate directly into eachunit cell 106 because the silicon chip 104 is used in a back-illuminatedconfiguration—essentially the photons 112 enter through the backside ofthe silicon chip 104.

[0038] Packaging the device 40, obviously, differs from that of theretinal prosthesis test device 30. Packaging the permanent implantdevice 40 requires that the photons 112 be allowed to pass through thebackside of the device 40. This is a simple matter of eliminating theceramic carrier taught in the retina prosthesis test device 30. Thinningthe silicon chip 104 is necessary because of the need for good quantumefficiency. The packaging scheme for the permanent implant device 40 isshown in FIG. 7. Note that the silicon chip 104 can be thinned to a fewtens of microns so that the overall mass of the object is primarily thatof the NCG array 102 making it more amenable for surgical attachment tothe retina 114.

[0039] It will be noted that there is no need for any multiplexingfunctions in the permanent implant device 40, therefore the design ofthe chip 104 is much simpler. Also there are no ancillary electronics,however, the silicon chip 104 significantly differs from that previouslyset forth, as shown in FIG. 9. Although there are no multiplexingrequirements, there are two new requirements, Specifically, these areexternal power and command signals necessary to adjust the operation ofthe device 40. Transmitting power and signals onto the device 40 areimplemented with an inductively driven coil or antenna 116. The majoron-chip electronic adjustments needed are control of bias supplies 118and biphasic pulse generator 122 plus the standard digital electronics124 that supply timing for the simultaneous operation of the unit cell106 sequences. Again the operation of the device 40 is to collect chargein the storage capacitors of the unit cells. The on-chip power receiver126 provides conditioned power to operate all the on-chip electronics.The frame rate would be nominally 60 frames per second, but becausethere is no longer a need to be compatible with the RS-170 format, theframe rate could be adjusted to anything desired.

[0040] Packaging of the permanent implant device 40 is very demanding.Along with issues of biocompatibility is the question of devicelifetime. Permanent implants might need to operate for several decades.Similar requirements exist for other electronic implants such as cardiacpacemakers and cochlear prosthetics. The encapsulation of the permanentimplant device 40 is easier in one respect than that of the retinalprosthesis test device 30 there are no connecting cables to the device40. In the case of the latter device 30, encapsulation was not acritical issue because the duration of the experiments are typicallyless than one hour. Cables connected to any neural prosthesis aresubject to mechanical forces that over time can damage seals andultimately cause failures. Because the permanent implant device 40 iscompletely wireless (no cable connections), simple encapsulation shouldbe achievable with high integrity.

[0041] The specific teachings of this approach to neural implants is anew intra ocular device, and has several extremely important advantagesover any device taught by the prior art. First, the use of the NCGenables the creation of a curved surface allowing the positioning of theelectrodes in extremely close proximity to the retinal cells over alarge area. NCG also allows the creation of very small electrodes (onthe order of a micron) with very high densities (thousands of electrodesper square millimeter). Regarding the multiplexer for the retinalprosthesis test device 30, as well as the unit cell size for thepermanent implant device 40, the unit cell size can be made as small aspractical based on the latest microelectronic design rules.

[0042] The multiplexer technology taught in the retinal prosthesis testdevice 30 uses only a small number of electrical leads through the eyewall while allowing a high data rate to the retina. In essence, thisallows the input of image sequences to the retina, at high resolutionand rapid frame rates. The electrical leads may be fabricated on a smalland flexible microcable. This is critical to performing humanexperiments and testing. This is in comparison to directly coupling allstimulating electrodes to individual leads. For example, a simple 8×8test array when directly coupled to input electronics outside the eyewould require 64 input leads. The cabling and accompanying connectorsfor such a device becomes very cumbersome.

[0043] The devices taught here 30 and 40 basically solves the technicalproblems of an earlier concept envisioned by the research group at theWilmer Ophthalmological Institute working with North Carolina StateUniversity (NCSU). An approach similar to NCSU is also under study by anMIT/Harvard team. Other efforts are proceeding in the United States,Germany and Japan that build on the basic idea of stimulating retinalcells with a small number of electrodes on a microelectronic chip.However, none of these approaches addresses the difficult issue ofhigh-density electrodes in close proximity to the retina for achievingvery high-resolution imagery.

[0044] A second approach is to stimulate the retina with amicroelectronic chip from behind the retina, either replacing thediseased photoreceptors or positioning the chip just behind them. Athird approach is to skip the retina altogether and stimulate the visualcenter of the brain. In this approach, an array, with penetratingmicroelectrodes is positioned against a visual cortex. This involvesinvasive brain surgery through the cranium. From a surgical point ofview, the intra ocular approach is the least invasive.

[0045] Although this invention has been described in relation to anexemplary embodiment thereof, it will be understood by those skilled inthe art that still other variations and modifications can be affected inthe preferred embodiment without detracting from the scope and spirit ofthe invention as described in the claims.

What is claimed:
 1. A retinal prosthesis test device comprised of: anexternal image source producing a video image; a micro-cable forconducting the video through a patents eye wall; an electronic chip fordemultiplexing the video image into a two-dimensional (2-D) array ofunit cells; a nanochannel glass electrode array hybridized to said 2-Darray of unit cells with induim bumps and electrically connecting eachunit cell to adjacent neural tissue; and an external electronic circuitboard generating a biphasic pulse applied globally to the unit cellsthrough the micro-cable causing an electronic signal to be applied to ahuman eye wherein it is converted to an electrochemical signaltransmitted through human tissue within the eye to a patients opticnerve.
 2. A device, as in claim 1, wherein the nanochannel glasselectronic array is comprised of a hybridized to the microelectronicchip mounted on a ceramic chip carrier, all unit cells controlled by aseries of row shift registers and column shift registers for directingthe incoming video image to a predetermined unit cell.
 3. A permanentretinal implant device comprised of: a nanochannel glass electrode arrayreceiving photons through a human iris and converting said photons to atwo-dimensional (2-D) spatially discrete electrical signal residing inindividual unit cells; an electronic unit located externally on apatient for inductively transmitting electrical power and controlsignals to the electronic chip and nanochannel glass electrode arraywithin the patients eye; and a biphasic pulse generated with on-chipelectronic circuitry causing an electrical signal to be routed throughthe nanochannel glass electrode array and applied to adjacent retinaltissue in a human eye where it is converted into an electrochemicalsignal to be transmitted through retinal neurons within the eye to apatients optic nerve
 4. A device, as in claim 3, wherein the nanochannelglass electronic array is further comprised of an on-chip antenna forreceiving the inductively transmitted signals and applying themselectively to an array of unit cells, digital electronics forcontrolling the application of the electrical to the retinal tissue, anon-chip power receiver for applying power to the array of unit cells, abias voltage supply for operating the electronic chip, and a biphasicpulse generator for generating the biphasic pulse, as well as row andcolumn shift registers.